© Turkish Society of Radiology 2017. PURPOSE Occlusion time (OT) is an important factor in the treatment of pulmonary arteriovenous malformations (PAVMs) since it can lead to serious complications. The purpose of our study is to calculate the OT of Amplatzer vascular plug (AVP, St Jude Medical), and correlate it to the type of the device used (AVP or AVP 2) and the percent of device oversizing. Technical success rates and complications were also recorded. METHODS We retrospectively studied a total of 19 patients with 47 PAVMs who received percutaneous transcatheter embolization therapy using either AVP or AVP 2. We recorded the location, type, feeding artery diameter, AVP device used, and OT of each PAVM. We correlated the percent of device oversizing and the type of AVP with the OT. We also studied the rate of persistence of PAVM for both devices. RESULTS Forty-six (98%) of the PAVMs were simple. Device diameters ranged from 4.0–16.0 mm with device oversizing ranging between 14% and 120%. There was a statistically significant difference in the OT of AVP and AVP 2 (3 min 54 s vs. 5 min 30 s, P = 0.030). There was a weak positive correlation between OT and device oversizing for AVP (r=0.246, P = 0.324) and AVP 2 (r=0.261, P = 0.240). No major complications were identified. Immediate technical success rate was 100%. CONCLUSION The use of AVP 2, and increase in device oversizing were not associated with reduction in the OT of PAVMs. There was no reported difference in safety between the two devices, and no major complications were noted.